SDZ - II acupuncture device
Six (6) output channels
New IC circuitry, no cross over, no interference.
Continuous, intermittent, and dense-disperse (modulated) wave forms
Adjustable frequency : 1 ~ 100 Hz
Durable alligator clip wires
Timer 0 ~ 60 minutes
Output safety alert - all outputs must be turned off before power on the unit.
Channel 6 outputs
Output Pulse Amplitude 0 ~ 50V (on 500 ohm load), adjustable
Pulse frequency Continuous wave : 1 ~ 100 Hz, adjustable
Intermittent wave : continuous wave
Frequency adjustable in 1~100 Hz, break-wave time is 5 seconds, continuous
Wave time is 15 seconds.
Dense-disperse wave : spacing wave frequency is 1/5 of density frequency,
Density wave frequency adjustable in 5~100 Hz.
Pulse width 1 ms
Pulse wave un-symmetry two-direction pulse wave.
Wave form adjustable, dense-disperse, and intermittent
Power Source DC 9V (CR14, UM2), or AC adaptor
Output Jack 3.5 mm diameter jack
Unit Dimensions 260 x 190 x 110 (mm)